Since around 1970, cryogenic systems have required the use of cold compressors. The requirement appeared for two reasons, The first was reduce the pressure of liquid helium with the intention of realizing the stable operation of large superconducting magnets with the best superconducting ability by decreasing operating temperature. The other was to improve the reliability of helium refrigerator compressors by introducing a turbo-compressor with oil-free bearings. This paper describes the circumstances and particular development of cold compressors, requirements related to helium refrigeration systems and cold compressors and future prospects.
The wire motion in a superconducting magnet has been analyzed by means of the Monte Carlo method. A sufficient analysis of the instability originating from wire motion has not been made because wire motion has not been clarified at a singular area in the coil, such as a “cross over” position where one wire crosses over other wires in the lower layer. A winding compression test was also performed and the location of disturbances in the coil was determined by the acoustic emission method. It was clarified that, in the singular area of a coil, wire motion is easily generated. The experimental results were compared with those calculated. It was confirmed that the instability in a coil originates from wire motion mainly in the singular area.
To standardize the test method for the critical current (Ic) measurement of three-component (Nb-Ti/Cu-Ni/Cu) superconducting wires, we have extracted many problems such as quenching, which often occurs in the wires, and have established a new measuring method. It was found that adding tension corresponding to a 0.1-0.2% strain applied to the wires is effective for preventing quenching than using silicone grease for fixing the wires to a mandrel, and that we can measure Ic's for all magnetic fields without quenching by soldering the wire to an alloy mandrel. It was also found that no trouble occurs when common Cu/Nb-Ti wires are measured based on the Ic test method of three-component wires. Therefore it is favorable to fix the wires by tension rather than using silicone grease in the case of wires which are easily quenched. Round robbin tests were carried out to check the validity of the draft for the test method. As a result of checking the coefficients of variation (COV) concerning individual experimental results, it was confirmed that the COV's reduce less than 2% in most cases. Based on these results, the draft of a test method for three-componen